BC Glusteron 2% designed for rapid, high level disinfection and sterilization of medical devices, dental and surgical instruments, heat-sensitive equipment, rigid and flexible endoscopes.
- Stable dialdegyde containing composition
- pH neutral
- Ready to use
- No activation required
- High disinfection and sporocidal activity
- Suitable for all heat-sensitive dental and surgical instruments
- Non corrosive.
BC Glusteron 2% solution contains glutaraldehyde - 2%, buffered compounds, corrosion inhibitor, auxiliaries, perfume, dye, water.
2% - for high level disinfection and cold sterilization.
BC Glusteron 2% is ready-to-use and does not require dilution.
Clean all items to be sterilized with a detergent or a pre-disinfectant product BC Pre CleanUp. Open all hinged instruments and other items. Disassemble those instruments with sliding or multiple parts because the solution must contact all surfaces for sterilization to be achieved. Place all items in the soaking bath so that they completely submerged by the solution. Soaking bath should always be covered with a lid.
Contact time: High level disinfection – 10 min; sterilization - 60 min.
Carefully remove the instruments from the solution and brush if necessary to remove any remaining visible debris. Rinse thoroughly under purified water. Ensure that instruments with recesses and channels are covered without any bubbles. Dry instruments with a clean disposable paper towel. Pack the instruments and keep in aseptic conditions.
More details of range and time of action see in instruction for use.
Range of action
Bactericidial (including tuberculocidal), fungicidal (including Candida Albicans and Aspergilus niger), virucidal (including HIV-1,HBV and BVDV(surrogate of HCV), sporocidal (Bacillus subtilis).
Warning and precautions for use
- Harmful by inhalation and if swallowed.
- Causes burns.
- Avoid contact with eyes and skin. In case of contact with eyes, rinse immediately with plenty of water and seek medical advice.
-Operate in well-ventilated areas.
-Use face protective mask and gloves.
-Keep out of reach of children.
-Keep at temperature from +5°C to +35°C.
-Do not use beyond expiry date.
-Shelf life of ready for use solutions - 30 days.
-Store in clean and tightly closed containers.